WELCOME TO THE SUPPORTAL
FREQUENTLY ASKED QUESTIONS
What are the recommended battery levels for safe use of the AvaTrace?
We recommend changing the batteries at 6.6V. The fast flash on the battery LED indicator is triggered at 6.6V.
My AvaTrace has been stolen. Can I use GPS to locate it?
Unfortunately, not. The AvaTrace M80 is not a GPS-enabled device. It is also unlikely that the mobile network provider will provide any location information if the unit is still connecting to the network.
Why won't my unit connect to AvaNet?
This could be one of many issues; the most common being low battery power and/or poor network coverage.
If a local SIM card is installed correctly but cannot connect to the network, is the international SIM activated?
No. If a local SIM card is installed correctly (detected by the instrument) the international eSIM will not be activated automatically.
What is the noise floor of the AvaTrace?
The noise spectral density for the AvaTrace M80 is typically around 0.35 μV/√Hz with a geophone sensor. With the filter profile for SS 480 48 66 (frequency range from 5 to 300 Hz) the noise floor is about 6 μV RMS.
What does the arrow on the geophone represent?
The arrow on the top of the geophone represents the longitudinal axis. The arrow should always be visible when looking directly down on the sensor. This is good rule-of-thumb for checking correct sensor orientation.
What is the sensitivity of the Geophone sensor?
How do I know that the AvaTrace is connected to AvaNet and is logging data?
When the Start/Stop LED indicator flashes slowly the unit is sampling. Successful connection to AvaNet is displayed by solid Connect and signal strength LED indicators.
What is power saving mode?
To save battery power when the instrument is inactive it goes into a power saving mode. This occurs 12 hours after the measurement is disabled and no other activity is registered.
What does the AvaTrace measure exactly?
The AvaTrace M80 device is designed to measure vibration velocity in millimetres per second (mm/s). It is targeted toward ground vibration from sources such as construction, demolition and blasting works.
What is the 'Raw Waveforms' checkbox in the Instrument Panel (Channel)?
By default, the AvaTrace stores filtered waveforms according to the selected channel filter profiles. If the *Raw Waveforms* box is ticked, raw unfiltered waveforms are be stored. This may be used when the end user would like to post-process the raw waveform data to extract information such as Whole Body Vibration (WBV) stats. N.B. All alarms and analytics will still be based on the filtered data.
What does the 'Auto Disconnect' checkbox in the instrument panel do?
By default, an AvaTrace M80 will automatically disconnect after successsfully downloading new settings and/or uploading any current data. If the 'Auto Disconnet' checkbox is left unchecked, then the next time the units connects it will hold the radio connection indefinitely. This may be helpful in situations such as attended vibration monitoring. We highly recommend leaving 'Auto Disconnect' enabled for long-term use to improve battery power conservation.
What do the signal strength values in the log file read out represent?
The RSSI parameter is the Received Signal Strength Indicator, as reported by the radio module (using the encoding specified in 3GPP TS 27.007):
Why are settings in the Instrument Panel highlighted yellow or red?
The Instrument Panel contains all the settings associated with the selected AvaTrace M80 unit. When configuration settings do not match those currently loaded on the unit they will be highlighted in either yellow or red.
What is a Measuring Point Day (MPD)?
When an AvaTrace M80 unit connects to AvaNet a 'Measuring Point Day' (MPD) is registered. The unit does not have to be sampling for this to occur. Simply a connection is sufficient to accumulate an MPD. MDPs are used for billing purposes to determine the total AvaNet daily access fees for unit owners.
I'm thinking about purchasing an AvaTrace device. Are there any additional operational costs/subscriptions to use this product?
Yes. AvaTrace devices have two additional costs associated with their operation: mobile network connectivity and an access fee to the AvaNet web portal.
AvaNet (Measuring Points)
How do I return all true entries in an AvaNet column?
==true (alternative for all false entries is ==false)
How do I filter results in an AvaNet table?
Filtering tables is a great way to find the data you are looking for quickly. If you know the right command you may write it straight into the text box at the top of the relevant column. If not, click the ? next to this box to help generate the code required for your filter.
I am missing some older data in AvaNet. Is it still accessible?
Yes. To improve navigation speed and usability AvaNet hides older data. To see all data associated with that measurement point click full query in the menu above the table. To revert back to the reduced table, click partial query.
How do I change the number of displayed lines per page in table view?
Select the menu item ”Preferences” which you will find in the menu ”Settings”. In the window that opens, select the desired number of rows and then click ”Finish”. The number of rows is now saved for future logins.
How do I search for values within a specific time period?
Sometimes you may want to display measurements for a period of time regardless of the date, e.g. every day between 07:00 – 18:00. In this case, you can use the "Time of day” instead of the column ”Time” which includes dates. Normally, the column ”Time of day” isn’t displayed, but you may add it by clicking the ”select columns” above the table. To enter the desired time interval, click the question mark to the left of the search field column ”Time of day”. This opens the dialog for advanced options that make your job of building up the search string. For example, to find out the range of 07: 00 – 18:00, do the following: 1) Select the option ”not before,” and enter 7:00 into the text field and click the ”And” 2) Select the ”not after:” and enter 18:00 into the text field and click the ”And” You have the following search string in the text field: > = 07:00:00 && <= 18:00:00 Click ”Search” to execute the search. The dialog closes and the table is updated. You can see the search string in the search field for the column ”Time of day” If you want to limit the search to be between certain dates, include the column ”Time” and simply choose the desired dates.
Can I download waveform data for measured events?
Yes. Waveforms automatically become available through AvaNet for logged events, i.e. events that exceed both the Trigger Level and Waveform Transfer Threshold (set in the Instrument Panel).
How do I view additional columns in the table format?
AvaNet has a default display set for different pages. To add hidden columns to a view, click add columns in the menu bar above the table. In the pop-up window select a metadata category you would like to view and press Add to move it to the right-hand column. You can also rearrange the order of columns using the Move up and Move down buttons.
What is a Measuring Point Group?
When generating a 'Vibrations, triax and air blast' measuring point in project set up, AvaNet automatically generates three measuring points associated with a single location: Transverse, Vertical and Longitudinal along with a fourth 'group' point which is associated with the three directional points. For a measurement point assigned demo, the associated points will be would be labelled demo-tram, demo-long, demo-vert and demo (The latter being the group measurement point). A group measurement point can also be created manually. The measurements for which there is a waveform uploaded to AvaNet appear in this group. Note that it does not include all measurements above the trigger level unless they also have a waveform in AvaNet. For all measurements above the trigger level, a waveform will exist on the unit, but they will not be uploaded to AvaNet unless the waveform threshold has been exceeded. If a measurement has exceeded the trigger level but not the waveform threshold the waveform may still be requested and thus uploaded to AvaNet. This is not a fast process and is not recommended as standard practice. Once the waveform has been requested and thus exists in AvaNet the measurement will appear in the Group Point.
How do I move an instrument to a new measuring point?
If you want to move to a new measuring point within the same project, select the measuring point in the list shown in the channel tab in the instrument panel. If you want to move the instrument to a measuring point in another project, detach the measuring point in the current project, switch to the new project and attach the new measuring point.
Why is the Measuring Point Group table empty?
A Measuring Point Group contains Logged Event data (i.e. measurements that exceed both the channel Trigger Level and Waveform Transfer Threshold).
Why can I not add an instrument to a new project?
Instruments can only be added to new projects when no Measuring Points are currently attached to the unit.
How do I rectify a self-test error?
A self-test error occurs when there is an issue with the incoming signal from the geophone. The most common source is a misaligned geophone. Ensure that the vertical (up), and horizontal (lateral) axis of the geophone are aligned appropriately. For visual reference, the arrow on the geophone should always point in the longitudinal axis.Other checks that you can carry out include checking that the trans cable is connected to the trans input and the vertical cable is connected to the vertical input etc. Checking that the TNC connectors are fully screwed in. You may try un-connecting and reconnecting these points. Checking the number and location of connected cables corresponds with the active channels in AvaNet. Checking that the cable isn’t damaged. Please note that once a self-test error has been triggered, the warning in AvaNet will only be removed by a positive self-test. Self-tests however are usually completed after the unit has connected to AvaNet on measurement start, thus the results will not be pushed to AvaNet until there is a reason to connect again (a positive self-test error does not trigger connection). If the measurement button is not flashing fast there is no error associated with the incoming signal. If you wish to remove the warning message in AvaNet, simply press the connect button.
What is a 'self-test' and when does it occur?
A 'self-test' is the method by which the AvaTrace determines correct geophone sensor operation. A 'self-test' occurs when an AvaTrace commences sampling and then once a day, generally overnight, for each day the unit is active. A 'self-test' will also be carried out if a channel’s status changes from inactive to active whilst the unit is sampling.
Does the log file report each time a self-test error is conducted?
Once a measurement has started, the self-test is not completed until after the unit has finished connecting to AvaNet.
What is a DPU Error Code warning?
This error can be ignored. It is a warning relevant to the developers that gives them a feel for how the power is affecting the units. It does not affect the general operation of the unit.
Why don't I have an instrument error alarm group?
The most common error here is selecting the finish button within the project set up wizard before the alarm set up section. Ensure you keep clicking next until the end of the wizard. Another issue may be related to user permissions. Contact Resonate Systems to ensure you have the appropriate permissions.
Should I send an SMS-alarm directly from the instrument of via an alarm group in AvaNet?
An SMS-alarm directly from the instrument arrives faster compared to the one sent from the alarm group in the AvaNet. SMS from the instrument are sent directly through your local network whereas SMS from AvaNet must be sent to the cloud and then to your phone, thus the slightly longer delay. Alarms sent directly from the instrument are limited to two recipients. Whereas, SMS from AvaNet may be sent to multiple recipients. To add a recipient to an alarm group it is required that the recipient is a registered user of the AvaNet. To send an alarm directly from the instrument only requires a cellphone number.
How do I set up and start a measurement?
Just connect the microphone and plug in the EM2010 and measurement starts automatically. The monitor will continue logging sound levels until power is removed. Changing the measurement settings is simple too. The user interface to the EM2010 is through a web browser over Wi-Fi link, so you can connect using a laptop or smart phone. We know it can be difficult to make changes to configurations when equipment is already installed. So the EM2010 is different, there are no small buttons or screens and no complicated menus to look through when you are on site. Simply connect the the Wi-Fi network and log in using your favourite browser. You can download readings, calibrate the monitor and change the measurement set up. Then just reboot the system and your changes are applied.
What happens if the power goes off?
The EM2010 is fully automated so it will just start measuring again as soon as power comes back on. And what's more, we will know if your measurement site goes down. Our performance monitoring software checks to ensure that your measurement site is reporting on schedule, so if power goes off we will email you to let you know.
How do the communications and remote upload work?
The EM2010/30 comes with a GSM modem installed. You can either utilise an inbuilt international SIM that works on the local 3G/GSM network or you can install your own local SIM. Readings are automatically uploaded to an external database every day so that you can access everything online. The cloud based software lets you manage your measurement sites and analyse your data. Should you use the inbuilt international SIM there is no need for a phone contract or trying to configure network settings. The data costs are part of our pricing model and are covered by an annual payment.
How do I access the readings?
When you receive one of our sound level monitoring stations we provide a secure login to our website which gives access to all your readings. Using this platform you can access raw data files, plot graphs, export spreadsheets and set up automated analysis functions for processing your data.
How do I calibrate the EM2010/EM2030?
Calibration is simple and fast. Carefully remove the microphone from the weatherproof protection assembly and place a standard acoustic calibrator (1/2") on the end. Connect to the EM2010 over the Wi-Fi interface and select the calibration option. Press the start button and the system will run the calibration program. The system will present an updated calibration value which you can choose to apply or ignore. Once you apply a new value the system will restart and begin measuring with your new settings.
Are 15min alarms possible?
Yes. At present, 15-minute alarms and must be set up remotely by Sonitus Systems on request.
What is the battery life of the Sonitus Field Kit?
Sonitus Field Kit with Battery and Solar Panels, indefinite depending on solar exposure.
What is low power mode?
Lower power mode is a new software update available on all models from EM2030 onwards. This will switch off the Wi-Fi after a preset number of minutes. This results in about 20% power saving. In order to connect to the unit once WiFi has been switched off, the unit must be rebooted.
Will the Sonitus send me an Alert if it isn't logging data?
You can set up a 'connectivity alert' to send an email when then unit has not uploaded data after a set period of time. This alert can only be sent once a day.
When will the unit start logging once i turn it on?
Depending on what the logging interval is set to, the monitor begins logging on the next even slot. So for example, if you were to power on a monitor at 10:33, the behaviour would be as follows 30 min logging -> start at 11:00 15 min logging -> start at 10:45 10 min logging -> start at 10:40 5 min logging -> start at 10:35 1 min logging -> start at 10:34
How quickly after a breech will I receive an alert?
Noise limit breeches are calculated based on a period average. Data is uploaded and processed every 15min. Alerts are sent out a few minutes after the upload has been completed.
What is the audio option?
Audio capture is an additional package that allows audio recording for up to 1 minute on a sound based trigger. This allows further investigation into the source of high levels. This feature must be organised on purchase of the instrument and can not be activated remotely.
What is the frequency analysis option?
This is an additional feature that is available with the Sonitus package that enables 1/3 or 1/1 octave analysis. This feature may be activated remotely by the Sonitus team should you wish to purchase it as an add on package to your existing Sonitus Noise Logger.
When I turn the unit off for the day how do I get that days data if it only uploads at fixed times, once a day?
The monitor uploads every 15 minutes. So readings should be almost up to date whenever power is removed. As soon as power is restored the monitor will upload the latest results that are stored and then begin a new measurement.
Is it possible to transfer the data via any means other than GPRS?
Data transfer across the Wi-Fi link is not possible at the moment, but it is planned for next year. It is possible to add an Ethernet port, but it is a factory option and needs to be organised at the time of purchase.
Do I need a rain hat for the Sonitus EM2010/EM2030 Microphone?
The windshield is made of a high quality material that has been treated to wick rain water. No rain hat required.
What is the typical data usage of a unit?
A unit operating with 5min intervals uses less than 3Mb per day (no audio). 1 min interval will use slightly more. 100MB per month plan is generally sufficient. If you have audio actvated, 1GB per month is usually sufficient.
Is it possible to install a local 3G SIM into the unit?
Yes. You will be responsinle for your own data charges and the annual subscription fee will be reduced.
Can I access the WIFI user interface remotely?
The monitor needs to be on and you need to be within WIFI range which is approximately 100m.
Within the chart view of the data visualisation tool can I toggle different trends on and off?
Yes, simply click on the data label below the chart header to toggle the trend line for this data on or off. The data options include L90, L10, LAeq and LAFmax.
What happens if I do not pay my yearly subscription?
The monitor will still work as per normal however you will have no access to the online platform. The international SIM card will be deactivated if it is in use. If a local SIM card is in use it won't be able to communicate with the web platform. This means no post processing, alerts, or remote support access. If you wish to let your subscription lapse with the option of renewal in the future and you are not set up with a local SIM you must arrange to have the unit settings changed to accept local sims prior to your disconnection. You will need to send the unit back to Ireland if this is not done before communications with the international SIM have been cut. Without a communciations package the unit can still be used and the data accessed directly from the unit.
Are there alarming options via email and/or SMS?
Yes. You can set alarms on the "Monitor" page of the web platform. We refer to these as Limits and you have the option to receive an email or SMS alert from the webplatform when a limit is breached. SMS's are not sent directly from the unit. When you are on the monitor page just click on "Manage Limits" to set up these alerts. There is a link to a tutorial once you get to the limit page.
What is the sample rate of the audio recording?
96kHz sample rate, 16-bit audio
Is there a time limit on the audio recording?
1 minute maximum (30 seconds pre-event and 30-seconds post-event)
What is the noise floor of the EM2010/EM2030 unit?
The noise floor on this unit is 20dB(A).
Can we change settings via the Sonitus Systems Web platform?
Almost! This feature is currently under development and should be released soon. Currently, audio capture settings can be changed remotely and if you are in a sticky situation Sonitus Support can log into the instrument and manage any settings chnages you may need. Please contact us if you need to access your instrument remotely.
Can I import my noise data into AvaNet?
Yes. You can import Noise data into AvaNet or Vibration Data into the Sonitus Cloud. Please contact Resonate Systems if you would like to explore these datahubs further.
Is 33db-121dB dynamic range adjustable?
The linear range which we test is 30-120dB(A). This is not adjustable.
Is there an option to do 15min rolling averages?
Yes. Please contact Resonate Systems if you would like this set up.
Have you ever had a request for a criterion line on the real time data plot?
Yes, we do get requests for this. There are a couple of reasons we have never implemented it; For non-expert users it is confusing - the alerts are usually based on an LAeq average, but users expert an alert when a single measurement (LAeq or Lmax) shows a breach. Where there are different time frames or various limits the plot gets quite busy - we generally try to stay on the side of simplicity to cater for all users. The more functionality we open up the more complicated the platform gets. We can activate these types of features in the background for expert users but do not show it as standard.
Is it possible to Audio trigger off a an octave band peak rather than overall peak?
Not yet, it is on the table as a new feature for future releases. Please contact us if you would like any further information.
What is the maximum size of an audio recording?
At 96kHz, 16 bit audio and 1min max recordings: 1546kb/s ... 92.16Mb .. 11.52MBA 1 minute recording will be 11.52MB.
Can I mount the microphone to a tripod?
The bottom of the stainless steel microphone mount has a 1/4" thread which will fit onto a camera tripod. You can buy small adapters for the top of survey tripods for heavy duty installations. These maybe avaibale locally from any surveyor or geotechnical equipment provider.
What does the standard Data package include?
The standard data subscription package covers everything that the monitor and web system are expected to do for 12 months. Bundled into the fee are the SIM charges, access the cloud platform, SMS alerts, and any support needed to keep your projects running. We also do some performance monitoring on the equipment to detect any faults e.g. disconnected microphone or comms issues.
Is the battery life affected by the logging interval?
No. The major source of power consumption is the modem and Wi-Fi. Upload schedule has the biggest influence on battery power. Where audio capture is activated, this will also have a large impact on battery life.